The purpose of this study is to design and fabricate an anthropopathic abdominal phantom for accuracy evaluation of deformable image registration (DIR) algorithms in adaptive radiation therapy. The constructed deformable organs, including the liver, kidney, spleen and stomach, are made of mixture of polyvinyl chloride (PVC) and softener dioctyl terephthalate, while the rigid structures, i.e. vertebrae, are made of white cement. Relation between the PVC-softener blending ratio and organ CT number is studied, and three-dimensional printing technic is employed to create highly anthropopathic organs in terms of organ shape and density. Detailed steps for phantom construction, landmark point placement and choice of phantom ingredients and construction recipe are introduced. Preliminary results of the mechanical properties of the fabricated organs are also presented. The experimental results indicate that the constructed phantom has satisfactory elastic characteristics and close CT number with corporal organs, and can potentially be applied to simulate real abdominal organ deformation in geometric accuracy validation of DIR algorithms.
Kirby N., , Chuang C., and Pouliot J., A two-dimensional deformable phantom for quantitatively verifying deformation algorithms. Med Phys 38, 4583 (2011).
Graves Y.J., , Smith A., , Mcilvena D., , Manilay Z., , Lai Y.K., , Rice R., , Mell L., , Jia X., , Jiang S.B., and Cerviño L., A deformable head and neck phantom with in-vivo dosimetry for adaptive radiotherapy quality assurance. Med Phys 42, 1490 (2015).
Singhrao K., , Kirby N., and Pouliot J., A three-dimensional head-and-neck phantom for validation of multimodality deformable image registration for adaptive radiotherapy. Med Phys 41, 121709 (2014).
Kashani R., , Hub M., , Kessler M.L., and Balter J.M., Technical note: A physical phantom for assessment of accuracy of deformable alignment algorithms. Med Phys 34, 2785 (2007).
Kashani R., , Lam K., , Litzenberg D., and Balter J., Technical note: A deformable phantom for dynamic modeling in radiation therapy. Med Phys 34, 199 (2007).
Chang J., , Suh T.S., and Lee D.S., Development of a deformable lung phantom for the evaluation of deformable registration. J Appl Clin Med Phys 11, 3081 (2010).
Hungr N., , Long J., , Beix V., and Troccaz J., A realistic deformable prostate phantom for multimodal imaging and needle-insertion procedures. Med Phys 39, 2031 (2012).
Serban M., , Heath E., , Stroian G., , Collins D.L., and Seuntjens J., A deformable phantom for 4D radiotherapy verification: Design and image registration evaluation. Med Phys 35, 1094 (2008).
Lorensen W.E., and Cline H.E., Marching cubes: A high resolution 3D surface construction algorithm. SIGGRAPH Comput. Graph. 21, 163 (1987).
Yeo U.J., , Taylor M.L., , Dunn L., , Kron T., , Smith R.L., and Franich R.D., A novel methodology for 3D deformable dosimetry. Med Phys 39, 2203 (2012).
Niu C.J., , Foltz W.D., , Velec M., , Moseley J.L., , Al-Mayah A., and Brock K.K., A novel technique to enable experimental validation of deformable dose accumulation. Med Phys 39, 765 (2012).