Ultrasound diagnosis is a widely used medical tool. Among the various ultrasound techniques, ultrasonic imaging is particularly relevant. This paper presents an improvement to a two-dimensional (2D) ultrasonic system using measurements taken from perpendicular planes, where digital signal processing techniques are used to combine one-dimensional (1D) A-scans were acquired by individual transducers in arrays located in perpendicular planes. An algorithm used to combine measurements is improved based on the wavelet transform, which includes a denoising step during the 2D representation generation process. The inclusion of this new denoising stage generates higher quality 2D representations with a reduced level of speckling. The paper includes different 2D representations obtained from noisy A-scans and compares the improvements obtained by including the denoising stage.
Galloway R.L., , McDermott B.A., , Thurstone F.L., A frequency diversity process for speckle reduction in real-time ultrasonic images, IEEE Transactions Ultrasonic, Ferroelectric and Frequency Control 35 (1988), 45-49.
Loizou C.P., , Pattichis C.S., et al., Comparative Evaluation of Despeckle Filtering In Ultrasound Imaging of the Carotid Artery, IEEE Transactions Ultrasonic, Ferroelectric and Frequency Control 52 (2005), 1653-1669.
Zhanga J., , Lina G., , Wua L., , Wanga C., , Cheng Y., Wavelet and fast bilateral filter based de-speckling method for medical ultrasound images, Biomedical Signal Processing and Control 18 (2015), 1-10.
Fu X., , Wang Y., , Chen L., , Tian J., An image despeckling approach using quantum-inspired statistics in dual-tree complex wavelet domain, Biomedical Signal Processing and Control 18 (2015), 30-35.
Dahdouh S., , Dubois M., et al., A 1D wavelet filtering for ultrasoundimages despeckling. Proc of SPIE 7629 (2010), 1-9, DOI: 10.1117/12.844388.
Nakayama K., , Ohyama W., et al., Speckle reduction of echocardiogramsvia wavelet shrinkage of ultrasonic RF signals. 13th ICBME 23 (2009), 395-398.
Zhang Y., , Wang Y.Y., et al., Doppler ultrasound signal denoisingbased on wavelet frames, IEEE Transactions Ultrasonic, Ferroelectric and Frequency Control 48 (2001), 709-716.
Liu B., , Wang Y., , Wang W., Spectrogram enhancement algorithm: A soft thresholding-based approach, Ultrasound Med Biol 25 (1999), 839-846.
Donoho D.L., , Johnstone I.M., Ideal spatial adaptation by waveletshrinkage, Biometrika 81 (1994), 425-455.
Donoho D.L., , Johnstone I.M., et al., Wavelet shrinkage: Asymptotia, J R Stat Soc B 57 (1995), 301-369.
Donoho D.L., , Johnstone I.M., Adapting to unknown smoothnessvia wavelet shrinkage, J Am Stat Assoc 90 (1995), 1200-1224.
Wang F., , Ji Z., Application of the Dual-tree ComplexWaveletTransform in Biomedical Signal Denoising, Bio-Medical Materials and Engineering 24 (2014), 109-115; DOI: 10.3233/BME-130790.
San Emeterio J.L., , Rodriguez-Hernandez M.A., Wavelet denoising of ultrasonic A-scans for detection of weak signals, 19th International Conference on Systems, Signals and Image Processing (IWSSIP), Vienna, Austria (2012), 54-57.
Coifman R.R., , Donoho D.L., Translation-invariant de-noising, Lect Notes Stat 103 (1995), 125-150.
Pardo E., , San Emeterio J.L., et al., Shift invariant wavelet denoisingof ultrasonic traces, ActaAcust United Ac 94 (2008), 685-693.
San Emeterio J.L., , Rodriguez-Hernandez M.A., Wavelet Cycle Spinning Denoising of NDE Ultrasonic Signals Using a Random Selection of Shifts, Journal Nondestructive Evalaluation 34 (2015), 270, DOI: 10.1007/s10921-014-0270-8. %
Rodriguez-Hernandez M.A., , %San Emeterio J.L., %Noise Reduction Using Wavelet %Cycle Spinning. Analysis of Useful Periodicities in the z-Transform Domain, %Signal Image Video Processing, in press, DOI: 10.1007/s11760-015-0762-8.
Rodriguez-Hernandez M.A., , San Emeterio J.L., Noise Reduction Using Wavelet Cycle Spinning. Analysis of Useful 189 Periodicities in the z-Transform Domain, Signal Image Video Processing, 10 (2016), 519-526; DOI: 10.1007/s11760-015-0762-8.
Meyer A.W., , Candy J.V., Iterative processing of ultrasonic measurements to characterize flaws in critical optical components IEEE Trans. Ultrason. Ferroelect. Freq. Control 8 (2002), 1124-1138.
Reguieg D., , Padilla F., et al., Ultrasonic transmission device based on crossed beam forming, Proc. IEEE Ultrasonics Symp. (2006), 2108-2111.
Zhu H.J., , Yang P., , Yao T.B., 3D reconstruction of ultrasound scanned data for tissue mimicking material sample, Bio-Medical Materials and Engineering 24 (2014), 2771-2781, DOI: 10.3233/BME-141095.
Rodriguez-Hernandez M.A., , Ramos A., , San Emeterio J.L., Wavelet-based 2D fusing of ultrasonic pulse-echo traces measured from two arrays radiating orthogonal beams, Measurement Science and Technology 22 (2011), 1-14, DOI: 10.1088/0957-0233/22/10/105701.
Rodriguez-Hernandez M.A., , Ramos A., , San Emeterio J.L., Localization of Isolate Flaws by Combination of Noised Signals Detected from Perpendicular Transducers, NDT and E International 37 (2004), 345-352.
Donoho D.L., De-noising by soft thresholding, IEEE Transactions on Information Theory 41 (1995), 613-627.